FORUM about all Other tube testers, not listed below
2 posts • Page 1 of 1
If the discrepance is very little, it may have to do with the tester also. Sometimes other scales switch some things in the circuit too.gilo wrote:I bought a Taylor 45A.
How to fix a taylor 45A?
The scales of 3mAV and 15mAV have discrepant readings.
Any problem with the tester, begins with checking the panel meter separated from the tester. You should leave it inside, but take of the wires. Then see if the full scale works good, and of linearity is good. If there is any stick effect etc.